## Generator Arbitrary

## signal generator

## Signal Generator

## Signal Generator Arbitrary waveform generator

## Waveform Pulse DDS Signal Generator

## DDS Arbitrary

## Signal Source Generator Arbitrary Waveform Frequency Meter

## Waveform Function Signal Generator Counter Kit

## DDS Signal

## Generator

## Arbitrary

## waveform generator Function signal generator

## DDS Arbitrary Waveform Function

## Function Arbitrary Waveform Pulse

## DDS Function Signal

## Arbitrary waveform

## Arbitrary Waveform

## Arbitrary sine Waveform

## Signal

## DDS Function

## Function Signal

## Digital DDS

## High Precision

## High Precision Digital DDS

## DDS

# DDS Signal Generator Arbitrary862 results

## DDS Signal Generator Arbitrary Product Descriptions

#### KKmoon High Precision Digital DDS Dual-channel Signal Source Generator Arbitrary Waveform Frequency Meter 200MSa/s 25MHz

Can output arbitrary waveforms and conventional function waveforms(sine/triangle/square/sawtooth wave and TTL signal)..

All functions are digitally controlled, precise and stable..

Dual-channel output, can work in sync, phase difference is adjustable..

With linear sweep(Max. up to 999s) and logarithmic frequency sweep functions..

Signal of both channels has built-in sophisticated -20dB attenuator in order to reach the minimum amplitude resolution 1mV..

Specifications

Normal Modes(Optional): 0.01Hz~25MHz

Square Wave: 0.01Hz~6MHz

Triangle Wave: 0.01Hz~6MHz

Sawtooth Wave: 0.01Hz~6MHz

Arbitrary Waveform: 0.01Hz~6MHz

TTL Digital Signal Wave: 0.01Hz~6MHz

Output Modulation: Frequency Sweep

The Min. Frequency Resolution: 10mHz

Amplitude Range(Peak to Peak

Offset Range: -120%~120% (The ratio of bias voltage and signal amplitude)

Bias Resolution: 1%

Phase Range: 0~359°

Phase Resolution: 1°

Sine Wave: Harmonic Arrived System 40dBc(

#### FY2300H 25MHz FeelTech Dual Channel DDS Function/Arbitrary Waveform Signal Generator 250MSa/s Sampling Rate with Intuitive Interface

FY-Economic series adopt direct digital synthesis (DDS) technology and FPGA design. With 14 kinds of frequently-used waveform internal preinstalled, it can also output arbitrary 8-bit 256 dots user-defined waveform..

It is easy to use with stable signal and low distortion..

Compact size, multiple functions, easy to carry..

It has 1 watt output, broadband frequency measurement and counter functions. The amplitude and DC offset of signal output are adjustable..

It's the ideal instrument for electronic engineering, laboratories, production lines, teaching and scientific research. It can also be used as supporting module for industrial equipment..

Function features:

- 3 downloadable 256 dots arbitrary waveform memories.

- 14 kinds of frequently-used waveform internal preinstalled. 3 positions are available for user-defined waveform.

- 1 watt signal output.

-The output waveform synthesized by function calculation. High accuracy and low distortion.

- Several different waveforms: Sine wave, Square wave, Triangle wave, Sawtooth wave, user-defined waveform, frequently-used waveform such as pulse, noise, and electrocardiogram etc.

- 10 sets of parameters defined by the users can be saved and loaded anytime.

- Operation mode: Button controlled, LCD1602 display, digital setting, knob adjusted continuously.

- Frequency of internal / external signal can be measured through built-in 20MHz frequency meter.

Main technology indexes

- Output waveforms: Sine wave, Square wave (Duty cycle adjustable) and Triangle wave, Sawtooth wave, frequently-used waveform such as pulse, noise, and electrocardiogram etc.

- Output amplitude: 0.5~8Vp-p (no load)

- Output impedance: 10Ω±10%(signal output)

- DC offset: ±3V

- Frequency range: 0.01Hz ~ 50KHz

- Resolution: 0.01Hz(10mHz)

- Frequency accuracy: ±5×10－6

- Frequency Stability: ±2×10－6/3 Hours

- Triangle linearity: >=98% (0.01Hz~5kHz)

- Sine wave distortion: less than 0.8% (reference frequency is 1kHz)

- Rise and fall time of square wave: ≤3.8us

- Square Wave Duty range: 1%~99%

- Counter Range: 0-65535

- Frequency Meter Range: 1Hz~20MHz

- Input Voltage Range: 0.5Vp-p~20Vp-p

- Display : LCD1602 in English

- Save and Load Parameter: M0-M9(M0: default load)

- Size: 83mm (Length)×60 mm(Width)×24mm(Height)

- Humidity:

#### iMeshbean High Precision Digital DDS Dual-channel Signal Source Generator Arbitrary Waveform Frequency Meter 25Mhz

Can output arbitrary waveforms and conventional function waveforms(sine/triangle/square/sawtooth wave and TTL signal)..

Dual-channel output, can work in sync, phase difference is adjustable.4 TTL output(phase difference is adjustable)..

With linear sweep(Max. up to 999s) and logarithmic frequency sweep functions. With USB interface, after the connection with PC, the PC can be used to control the instrument..

Signal of both channels has built-in sophisticated -20dB attenuator in order to reach the minimum amplitude resolution 1mV..

Has a frequency measurement, period measurement, positive and negative pulse width measurement, duty cycle measurements and counting function..

Package List: 1 * Signal Source Generator 1 * USB Cable 1 * CD 1 * TTL Expansion Board 2 * BNC Clips Cable

#### Jinwen 24MHz Dual-Channel Arbitrary Waveform DDS Function Signal Generator Kit FY3200S

Output signal frequency high accuracy, good stability.

Signal amplitude up to 20Vpp; Pulse wave pulse width precision can be adjustable.

Instrument can realize the adjustment program of function parameters , easy to use.

Accompanying frequency meter, counter function, high cost.

The instrument can output sine, square, triangle, ramp, pulse, staircase, DC voltage waveform.

Function:

◆ using DDS direct digital synthesis technology to generate precise, stable, low distortion output signal; ◆ CH1 and CH2 completely symmetrical two channels, each channel can be independently set the parameters.

◆ supports user-defined waveforms, each arbitrary waveform memory depth 2048 * 12bits, 250MSa / s sampling rate;

◆ has burst burst output, selectable manual trigger, internal and external trigger CH2 trigger three trigger modes that allow the unit to any output 1 ~ 1048575 bursts;

◆ measurement functions: instrument comes with 100MHz frequency counter functions and pulse counter functions

◆ waveform generator up to 12 bits wide, the output waveform is more delicate, waveform distortion is reduced magnitude

Technical Parameters:

Built-in switching power supply: AC85V-AC260V

Interface: USB

Signal output:

Output amplitude: ≥≥20Vp-p (no load)

Output Impedance: 50Ω ± 10%

DC polarization: ± 10V

Frequency range: 0Hz - 24MHz (orther 0Hz - 6MHz)

Counter:

Count range: 0-4294967295

Measurement frequency range: 1Hz~100MHz

Input amplitude: 1Vp-p~20Vp-p

Scan function:

Scanning mode: Linear sweep, logarithm scanning Frequency Range: Start and end points set arbitrary Frequency scanning range: M1 ~ M2 predetermined frequency

Scan rate: 1s ~ 999s / step

Package include:

1 x 24MHz Dual Channel DDS Function Signal Generator

1 x 1m power cable

1 x 1.25mUSB cable

2 x 1.1m Q9 cables

1 x disc

#### Wisamic 24MHz Dual Channel DDS Function Signal Generator, High Precision Arbitrary Waveform Counter

Wisamic Arbitrary Waveform Generator adopts large scale CMOS integrated circuit and high speed microprocessor. The internal circuit adopts active crystal oscillator as benchmark. So the signal stability is greatly strengthened..

Using Dual-channel DDS direct digital synthesis technology to generate precise, stable, low distortion output signal. includes Sine wave, Square wave, Triangle wave, Saw tooth wave and user-defined waveform, each channel can be independently set the parameters..

It has TTL electric level output, External frequency measurement, counter and sweep functions including Linear sweep and Logarithmic sweep. Both the sweep frequency and time can be set arbitrarily..

Storage feature: You can store 20 groups instrument state parameters set by the user, can be called up to Reproduce..

This Signal Generator is the ideal instrument for electronic engineering, laboratories, production lines, teaching and scientific research. The DC bias level is adjustable from -10V to + 10，other similar product has no this function..

Features

√High frequency accuracy: frequency accuracy up to 10-6 magnitude.

√High frequency resolution: the full range of frequency resolution 10mHz.

√Tracking function: built-in frequency, amplitude, offset, duty cycle, waveform and other parameters follow function, easy to use.

√Scanning features: a frequency linear sweep and logarithmic sweep function, scan start and end points can be set.

√Storage feature: You can store 20 groups instrument state parameters set by the user, can be called up to Reproduce.

√Operation: key operation, LCD1602 LCD display in English, direct digital setting parameters or continuous adjustment knob.

√Input voltage protection: extended power input range AC85V to AC240V wide voltage.

√Output short circuit protection: All signal outputs can operate at over load short circuit 60S. Specification Signal output:

Output amplitude: ==20Vp-p (no load).

Output Impedance: 50O ± 10%.

DC polarization: ± 10V.

Frequency range: 0Hz - 24MHz (orther 0Hz - 6MHz). Counter:

Count range: 0-4294967295

Measurement frequency range: 1Hz~100MHz

Input amplitude: 1Vp-p~20Vp-p Scan function:

Scanning mode: Linear sweep, logarithm scanning

Frequency Range: Start and end points set arbitrary

Frequency scanning range: M1 ~ M2 predetermined frequency

Scan rate: 1s ~ 999s / step Other:

Storage and transferred functions: M0 ~ M9 (M0: default)

Size: 200mm (L)×190mm(W) ×90mm(H)

Beeper function: to set open or closed by the program Package content

1x 24MHz Dual Channel DDS Function Signal Generator

1x 1m Power cable

1x 1.25m USB cable

2x 1.1m Q9 cables

1x Disc

#### Kuman 30MHz High Precision Dual-channel DDS Arbitrary Signal Waveform Generator Counter, 2.4in Screen Display,250MSa/s, 819214bits,Frequency meter, VCO, Burst, Modulation Function FY6600

Adopt the Direct Digital Synthesizer (DDS) technology and provide stable, precise, pure and low distortion signals.

2.4 inch TFT Color LCD with 320×240 resolution, displaying parameters and graphics of the two channels at the same time.

14-bit high-speed D/A converter chip (5Vpp output quantization error is less than 1mV), 250MSa/s sample rate, 14bits vertical resolution.

Various modulation types: AM, FM, PM, ASK, FSK and PSK modulations.

Various function types: weep, VCO (Voltage Control Output), Burst Output, 100M Frequency meter, Counter, Channel SYNC, Save, and Communicating..

Features

Various waveforms: Sine, Square (Duty Cycle adjustable), Pulse (Pulse width and cycle time can be set accurately), Triangle/Ramp, CMOS(0~10V), Four channels TTL, Exponential Rise, Exponential Fall, Noise, ECG, DC etc

Storage:Enable to store 64 arbitrary waveform data files, each one of waveform storage depth 8192 points * 14bits

Support two or more signal generators connected to achieve multi-channel output, the maximum support 16-channel synchronous output, the phase between each channel can be adjusted

Output short-circuit protection: All channels can work more than 60 seconds when the load is short-circuited

Advantages

5Vpp output quantization error is less than 1mV

Minimum amplitude resolution can be up to 1mV

Duty cycle of each channel can be adjusted separately. Precision can be 0.1%

Frequency Characteristic

Sine:0~30MHz

Square Wave: 0~25MHz

Ramp, Triangle, Pulse, Arbitrary Waveform: 0~10MHz

TTL/CMOS:0~20MHz

Frequency accuracy: ±20ppm

Frequency stability: ±1ppm/ 3hours

Minimum pulse width:20ns

Output characteristics

Amplitude Resolution:1mV

Amplitude Stability: ±0.5%/ 5 Hours

Amplitude flatness: ±2.5%(10MHz)

Frequency Meter Function

Measurement Objects: Frequency, Period, Positive/Negative Pulse Width, Duty Cycle

Input Voltage Range: 2Vpp~20Vpp

Frequency Meter: (Resolution: 0.01Hz; Range: 0.01Hz~100MHz; )

Counter Function

Counting region: 0-4294967295

Coupled mode: 2kinds Coupling modes, DC and AC

Counting mode: Manual operation

Package List

1 * DDS Signal Generator

1 * Power Cable

1 * USB Data Cable

2 * BNC-Clip Cable (Signal output cable)

1 * BNC-BNC Cable (Signal output cable)

1 * CD (Contain the user manual and PC software)

1 * Warranty Card

#### KKmoon High Precision Digital DDS Dual-channel Signal Source Generator Arbitrary Waveform Frequency Meter 200MSa/s 6MHz

Can output arbitrary waveforms and conventional function waveforms(sine/triangle/square/sawtooth wave and TTL signal)..

All functions are digitally controlled, precise and stable.With 120% ~ + 120% DC bias function..

Dual-channel output, can work in sync, phase difference is adjustable.4 TTL output(phase difference is adjustable)..

With linear sweep(Max. up to 500s) and logarithmic frequency sweep functions.Has a frequency measurement, period measurement, positive and negative pulse width measurement, duty cycle measurements and counting function..

Signal of both channels has built-in sophisticated -20dB attenuator in order to reach the minimum amplitude resolution 1mV.With USB interface, after the connection with PC, the PC can be used to control the instrument..

Specifications

Normal Modes: 0.01Hz~6MHz

Square Wave: 0.01Hz~6MHz

Triangle Wave: 0.01Hz~6MHz

Sawtooth Wave: 0.01Hz~6MHz

Arbitrary Waveform: 0.01Hz~6MHz

TTL Digital Signal Wave: 0.01Hz~6MHz

Output Modulation : Frequency Sweep

The Min. Frequency Resolution : 10mHz

Amplitude Range(Peak to Peak)

Offset Range : -120%~120% (The ratio of bias voltage and signal amplitude)

Bias Resolution : 1%

Phase Range : 0~359°

Phase Resolution : 1°

Sine Wave : Harmonic Arrived System ,40dBc(

#### KKmoon 40MHz Function Signal Generator Dual-channel DDS Arbitrary Waveform Pulse Signal Generator 1Hz-100MHz Frequency Meter

Adopt 320x240 2.4 inches TFT color LCD display, simultaneously display parameters of CH1 and CH2.

Ability to reproduce patterns of custom waveform..

Use soft silicone keys instead of traditional patch keys, high sensitivity and convenient for operation..

Using DDS direct digital synthesis technology to generate precise, stable and low distortion output signal..

Max. output frequency(sine wave) up to 15MHz/30MHz/40MHz(optional), 200MSa/s sampling rate and 14bits vertical resolution..

Features:

Using DDS direct digital synthesis technology to generate precise, stable and low distortion output signal.

Max. output frequency(sine wave) up to 15MHz/30MHz/40MHz(optional), 200MSa/s sampling rate and 14bits vertical resolution.

Completely independent dual channels output, synchronizing operation and adjustable phase difference.

Great improvement for square wave performance, square frequency range up to 15MHz with no distortion.

60 groups of arbitrary wave storage function, memory depth of each group is 2048*14bits.

High frequency accuracy and resolution: frequency accuracy up to 10^(-6) magnitude, total range frequency resolution is 0.01uHz.

High amplitude resolution: the minimum amplitude resolution can reach 1mV, amplitude range is 0Vpp~20Vpp.

Duty-cycle of each channel can be adjusted independently, the adjusting resolution is 0.1%.

Sweep function: linear sweep and logarithmic sweep functions; 0.1s-999.9s sweep time; sorward, backward and roundtrip sweep directions.

With pulse trains burst output function: allow the unit output 1~1048575 arbitrary pulse trains.

Measurement functions: can measure signal frequency, cycle, positive/negative pulse width and duty cycle; 1Hz~100MHz frequency range.

-9.99V~9.99V DC offset range, offset resolution: 0.01V.

With digital signal output function, CMOS output range 0~10V.

Storage feature: can store 100 groups of parameters set by the user and extract at any time.

Counter function: with DC and AC coupling modes.

Arbitrary wave editing function: user can edit arbitrary wave on PC and then download to output from the instrument.

Communication function: the PC can be used to control the instrument.

With output short-circuit protection: all signal output can work 60s under load short-circuit condition.

#### kuman 24MHz Dual-channel DDS Function Signal Generator, High Precision Arbitray Waveform Generator FY3224S

Kuman Arbitray Waveform Generator adopts large scale CMOS integrated circuit and high speed microprocessor. The internal circuit adopts active crystal oscillator as benchmark. So the signal stability is greatly strengthened..

Using Dual-channel DDS direct digital synthesis technology to generate precise, stable, low distortion output signal. includes Sine wave, Square wave, Triangle wave, Saw tooth wave and user-defined waveform, each channel can be independently set the parameters..

it has TTL electric level output, External frequency measurement, counter and sweep functions including Linear sweep and Logarithmic sweep. Both the sweep frequency and time can be set arbitrarily..

Storage feature: You can store 20 groups instrument state parameters set by the user, can be called up to Reproduce..

This Signal Generator is the ideal instrument for electronic engineering, laboratories, production lines, teaching and scientific research. The DC bias level is adjustable from -10V to + 10V ，other similar product has no this function.

Kindly remind:

Before using this instrument, please check if the power supply is normal, to ensure the normal use and personal safety

This instrument must be used in the technical index range.

Please do not change the instrument circuit arbitrarily, so as to avoid damaging equipment or endangering the safety

Function Features:

Sampling rate up to 250 MSa/s.

Built-in arbitrary waveform with 250 MSa/s sampling rate.

4 downloadable 2048 dots arbitrary waveform memories

With 12 bit wide waveform generator, the output waveform can be more delicate with low distortion.

Each function can be adjusted by host computer.

Preinstalled 17 common waveforms.

High resolution: Full range frequency resolution can be 10 mHz.

Both main and subsidiary wave duty cycle are adjustable separately (0.1%~99.9%） .

All range continuously adjustable, digital directly setting.

High waveform accuracy: The output waveform synthesized by function calculation is of high accuracy and low distortion.

Arbitrary waveform: User can load arbitrary waveform according to the need.

Sweep Function： Linear sweep and Logarithmic sweep. Starting and stop points can be set optionally.

Package Contents:

1x 24MHz Dual Channel DDS Function Signal Generator

1x 1m power cable

1x 1.25mUSB cable

2x 1.1m Q9 cables

1x disc

#### DDS Function Signal ,Generator Module, Awakingdemi DIY DDS Function Signal Module Sine Square Sawtooth Triangle Wave Kit

DDS Signal Generator provide lots of electronic components for you to do it by yourself..

Simple circuit with easily accessible and good components..

Delicated high speed (HS) signal output up to 8MHz..

DDS signals: sine, square, saw, rev saw ,triangle, ECG and noise..

Has two outputs-one for DDS signal and another for high speed aquare signal..

Features:

Provided lots of electronic components for you to do it by yourself.

Simple circuit with easily accessible and cheap components.

Delicated high speed (HS) signal output up to 8MHz.

DDS signal with variable amplitude and offset.

DDS signals: sine, square, saw, rev saw ,triangle, ECG and noise.

2 * 16 LCD menu, intuitive 5 button keypad.

Frequency adjusting steps: 1, 10, 1000, 10000MHz.

Restoring last configuration after power up.

Has two outputs-one for DDS signal and another for high speed aquare signal.

Specifications:

Material: PCB + Electronic Components

Power Supply: +5V / +12V / -12V

Menu Displey: 2 * 16 LCD

HS Signal Frequency: 1, 2, 4, 8MHz

Red Board Size: 8 * 8cm / 3.15 * 3.15in (L * W)

Red Board Weight: 20g / 0.7oz

Screen Size: 6.5 * 1.5cm / 2.6 * 0.6in (L * W)

Screen Board Weight: 29g / 1.04oz

Package Size: 15 * 11 * 2cm / 5.9 * 4.3 * 0.8in (L * W * H)

Package Weight: 101g / 3.58oz

Package List:

1 * Bag of Electronic Components