## JDS6600 Series

## JDS6600 Generator

## JDS6600 Arbitrary Sine

## JDS6600 Function Signal

## JDS6600 Frequency Meter

## JDS6600 Digital Control

## JDS6600 Signal Generator

## JDS6600 Arbitrary Waveform

## JDS6600 Generator Arbitrary

## JDS6600 Series 40MHZ Digital

## JDS6600 Meter Arbitrary Sine

## DDS

## JDS6600 Series Digital Control

## JDS6600 Arbitrary sine Waveform

## JDS6600 Signal Generator Counter

## JDS6600 Waveform Signal Generator

## JDS6600 Series 60MHZ Digital Control

## Series 60MHZ

## JDS6600 Signal Generator Arbitrary Waveform

## JDS6600 Pulse Signal Source Frequency Meter

## JDS6600 Signal Generator frequency meter Arbitrary

## DDS Signal

## Channel DDS

## DDS Function

## JDS6600 DDS Function Signal Generator Frequency Meter Arbitrary

## 60MHZ Digital

## Channel DDS Signal

## Signal

## DDS Signal Generator

## Pulse Signal

## Dual Channel

## Digital Contro Dual Channel DDS

## Signal Source

# JDS6600240 results

## JDS6600 Product Descriptions

#### SainSmart JDS6600 Dual-Ch DDS Function Arbitrary Waveform Signal Generator 60MHZ CNC

Using DDS direct digital synthesis technology to generate precise, stable and low distortion output signal.

Use soft silicone keys instead of traditional patch keys, high sensitivity and convenient for operation.

Max. output frequency(sine wave) up to 15MHz/30MHz/40MHz/50MHz/60MHz(optional), 266MSa/s sampling rate and 14bits vertical resolution.

60 groups of arbitrary wave storage function, memory depth of each group is 2048*14bits.

High frequency accuracy and resolution: frequency accuracy up to 10(-6) magnitude, total range frequency resolution is 0.01uHz.

Features:

*Arbitray Waveform Generator adopts large scale FPGA integrated circuit and high speed MCU microprocessor. The internal circuit adopts active crystal oscillator as benchmark. So the signal stability is greatly strengthened.

*Using Dual-channel DDS signal and TTL electric level output to generate precise, stable, low distortion output signal. includes Sine wave, Square wave, Triangle wave, Saw tooth wave, Pulse wave, white noise, user-defined waveform etc. each channel can be independently set the parameters.

*With linear sweep(Max. up to 999.9s) and logarithmic frequency sweep functions.Has a frequency measurement, period measurement, positive and negative pulse width measurement and counting function.

*Storage feature: You can store 99 groups instrument state parameters set by the user, can be called up to Reproduce. Frequency output of Sine wave can be up to 60MHz. 200MSa/s sampling rate. It has 60 positions for saving user-defined waveform. Waveform Length of each one is 2048 and vertical resolution is 14 bits.

*This Signal Generator is the ideal instrument for electronic engineering, laboratories, production lines, teaching and scientific research.

Specification:

*Sine Wave: 0~60MHz

*Square & Triangle Wave: 0~15MHz

*Pulse & Arbitrary & TTL Digital Signal Wave: 0~6MHz

*Square rise time: ≤20nS

*Frequency accuracy: ±20ppm×10-6

*Frequency stability: ±1ppm×10-6/3h

Packing List:

1 x DDS Signal Generator

1 x USB Cable

1 x Power Adapter

2 x Q9 Cable

1 x CD

#### JDS6600 Series 40MHZ Digital Contro Dual Channel DDS Signal Generator/ Pulse Signal Source Frequency Meter

Adopt 320*240 2.4" TFT color LCD display, simultaneously display parameters of CH1 and CH2，Ability to reproduce patterns of custom waveform..

60 groups of arbitrary wave storage function, memory depth of each group is 2048*14bits..

Duty-cycle of each channel can be adjusted independently0.1%-99.9%, the adjusting resolution is 0.1%..

Scanning features: linear sweep and logarithmic sweep functions; 0.1s-999.9s sweep time; sorward, backward and roundtrip sweep directions..

With output short-circuit protection: all signal output can work 60s under load short-circuit condition.

JDS6600 Series 40MHZ Digital Contro Dual Channel DDS Signal Generator/ Pulse Signal Source Frequency Meter

Features:

1.Use soft silicone keys instead of traditional patch keys, high sensitivity and convenient for operation.

2.Using DDS direct digital synthesis technology to generate precise, stable and low distortion output signal.

3.Max. output frequency(sine wave) up to 40MHz, 266MSa/s sampling rate and 14bits vertical resolution.

4.Completely independent dual channels output, synchronizing operation and adjustable phase difference.

5.Great improvement for square wave performance, square frequency range up to 15MHzwith no distortion.

6.High frequency accuracy and resolution: frequency accuracy up to 10^(-6) magnitude, total range frequency resolution is 0.01uHz.

7.High amplitude resolution: the minimum amplitude resolution can reach 1mV, amplitude range is 0Vpp~20Vpp.

8.Pulse pulse width and pulse cycle time is precisely adjustable, adjust the range of 30nS-4000S

9.With pulse trains burst output function, allow the unit output 1~1048575 arbitrary pulse trains.

10.Measurement functions: can measure signal frequency, cycle, positive/negative pulse width and duty cycle; 1Hz~100MHz frequency range.

11.-9.99V~9.99V DC offset range, offset resolution: 0.01V.

12.With digital signal output function, CMOS output range 0~10V.

13.Storage feature: can store 100 groups of parameters set by the user and extract at any time.

14.Counter function: with DC and AC coupling modes.

15.Arbitrary wave editing function: user can edit arbitrary wave on PC and then download to output from the instrument.

16.Communication function: the PC can be used to control the instrument.

Package:

1 x JDS6600 Series 40MHZ Digital Contro Dual Channel DDS Signal Generator/ Pulse Signal Source Frequency Meter

#### SainSmart JDS6600 Dual-Ch DDS Function Arbitrary Waveform Signal Generator 40MHZ CNC

High frequency precision up to 10-6 orders of magnitude..

Full range of 0.01μHz frequency resolution..

Minimum amplitude resolution can be up to 1mV..

Duty cycle of each channel can be adjusted separately. Precision can be 0.1%..

Features:

*Arbitray Waveform Generator adopts large scale FPGA integrated circuit and high speed MCU microprocessor. The internal circuit adopts active crystal oscillator as benchmark. So the signal stability is greatly strengthened.

*Using Dual-channel DDS signal and TTL electric level output to generate precise, stable, low distortion output signal. includes Sine wave, Square wave, Triangle wave, Saw tooth wave, Pulse wave, white noise, user-defined waveform etc. each channel can be independently set the parameters.

*With linear sweep(Max. up to 999.9s) and logarithmic frequency sweep functions.Has a frequency measurement, period measurement, positive and negative pulse width measurement and counting function.

*Storage feature: You can store 99 groups instrument state parameters set by the user, can be called up to Reproduce. Frequency output of Sine wave can be up to 40MHz. 200MSa/s sampling rate. It has 60 positions for saving user-defined waveform. Waveform Length of each one is 2048 and vertical resolution is 14 bits.

*This Signal Generator is the ideal instrument for electronic engineering, laboratories, production lines, teaching and scientific research.

Specification:

*Sine Wave: 0~40MHz

*Square & Triangle Wave: 0~15MHz

*Pulse & Arbitrary & TTL Digital Signal Wave: 0~6MHz

*Square rise time: ≤20nS

*Frequency accuracy: ±20ppm×10-6

*Frequency stability: ±1ppm×10-6/3h

Packing List:

1 x DDS Signal Generator

1 x USB Cable

1 x Power Adapter

2 x Q9 Cable

1 x CD

#### JDS6600 Series 60MHZ Digital Contro Dual Channel DDS Signal Generator/ Pulse Signal Source Frequency Meter

Adopt 320*240 2.4" TFT color LCD display, simultaneously display parameters of CH1 and CH2，Ability to reproduce patterns of custom waveform..

60 groups of arbitrary wave storage function, memory depth of each group is 2048*14bits..

Duty-cycle of each channel can be adjusted independently0.1%-99.9%, the adjusting resolution is 0.1%..

Scanning features: linear sweep and logarithmic sweep functions; 0.1s-999.9s sweep time; sorward, backward and roundtrip sweep directions..

With output short-circuit protection: all signal output can work 60s under load short-circuit condition.

JDS6600 Series 60MHZ Digital Contro Dual Channel DDS Signal Generator/ Pulse Signal Source Frequency Meter

Features:

1.Use soft silicone keys instead of traditional patch keys, high sensitivity and convenient for operation.

2.Using DDS direct digital synthesis technology to generate precise, stable and low distortion output signal.

3.Max. output frequency(sine wave) up to 60MHz, 266MSa/s sampling rate and 14bits vertical resolution.

4.Completely independent dual channels output, synchronizing operation and adjustable phase difference.

5.Great improvement for square wave performance, square frequency range up to 15MHzwith no distortion.

6.High frequency accuracy and resolution: frequency accuracy up to 10^(-6) magnitude, total range frequency resolution is 0.01uHz.

7.High amplitude resolution: the minimum amplitude resolution can reach 1mV, amplitude range is 0Vpp~20Vpp.

8.Pulse pulse width and pulse cycle time is precisely adjustable, adjust the range of 30nS-4000S

9.With pulse trains burst output function, allow the unit output 1~1048575 arbitrary pulse trains.

10.Measurement functions: can measure signal frequency, cycle, positive/negative pulse width and duty cycle; 1Hz~100MHz frequency range.

11.-9.99V~9.99V DC offset range, offset resolution: 0.01V.

12.With digital signal output function, CMOS output range 0~10V.

13.Storage feature: can store 100 groups of parameters set by the user and extract at any time.

14.Counter function: with DC and AC coupling modes.

15.Arbitrary wave editing function: user can edit arbitrary wave on PC and then download to output from the instrument.

16.Communication function: the PC can be used to control the instrument.

Package:

1 x JDS6600 Series 60MHZ Digital Contro Dual Channel DDS Signal Generator/ Pulse Signal Source Frequency Meter

#### JDS6600 Series Digital Control Dual-channel DDS Function Signal Generator Frequency Meter Arbitrary Sine Waveform (15MHz JDS6600-15M)

Using DDS direct digital synthesis technology to generate precise, stable and low distortion output signal.

60 groups of arbitrary wave storage function, memory depth of each group is 2048*14bits.

High frequency accuracy and resolution: frequency accuracy up to 10(-6) magnitude, total range frequency resolution is 0.01uHz.

Use soft silicone keys instead of traditional patch keys, high sensitivity and convenient for operation.

Max. output frequency(sine wave) up to 15MHz/30MHz/40MHz/50MHz/60MHz(optional), 266MSa/s sampling rate and 14bits vertical resolution.

Description:

Adopt 320*240 2.4" TFT color LCD display, simultaneously display parameters of CH1 and CH2.

Silicone button and jog dial,Completely change the previous patch button jerky feeling, in improving the service life and bring you a comfortable experience.

A small signal capable of outputting 2mVpp,The range resolution is 1mV,And you can do 1mVpp,10mVpp, 100MVpp and so on are differentStep value, amplitude range 2mVpp-20Vpp, stepless regulation.

Sinusoidal frequency range is0.01uHz~60MHz,The frequency range of square wave and triangle wave is0.01uHz~15MHz,The frequency range of other waveforms is0.01uHz~6MHzThe frequency accuracy can reach the magnitude of 10-6, and the full range frequency resolution0.01uHzThe accuracy of the output frequency is fully guaranteed;14 bit waveform vertical resolution, 266MSa/s waveform sampling rate, waveform more delicate.

Measurement functions: can measure signal frequency, cycle, positive/negative pulse width and duty cycle; 1Hz~100MHz frequency range.

Using DDS direct digital synthesis technology to generate precise, stable and low distortion output signal.

#### MagiDeal JDS6600 Series 30MHZ Digital ControL Dual-Channel DDS Function Signal Generator/ Pulse Signal Source Frequency Meter

JDS6600-30MHz dual-channel DDS Function Signal Generator, adopt 2.4inch TFT color LCD display, simultaneously display parameters of CH1 and CH2，ability to reproduce patterns of custom waveform..

Use soft silicone keys instead of traditional patch keys, high sensitivity and convenient for operation. Using DDS direct digital synthesis technology to generate precise, stable and low distortion output signal..

High frequency accuracy and resolution: frequency accuracy up to 10^(-6) magnitude, total range frequency resolution is 0.01uHz; Max. output frequency(sine wave) up to 30MHz, 266MSa/s sampling rate and 14bits vertical resolution..

Completely independent dual channels output, synchronizing operation and adjustable phase difference. Great improvement for square wave performance, square frequency range up to15MHz with no distortion..

Scanning features: linear sweep and logarithmic sweep functions; 0.1s-999.9s sweep time; forward, backward and roundtrip sweep directions..

Description:

- Adopt 2.4inch TFT color LCD display, simultaneously display parameters of CH1 and CH2，Ability to reproduce patterns of custom waveform.

- Use soft silicone keys instead of traditional patch keys, high sensitivity and convenient for operation.

- High frequency accuracy and resolution, low distortion output signal; frequency accuracy up to 10^(-6) magnitude, total range frequency resolution is 0.01uHz; Max. output frequency(sine wave) up to 30MHz, 266MSa/s sampling rate and 14bits vertical resolution.

- Completely independent dual channels output, synchronizing operation and adjustable phase difference.

- Scanning features: linear sweep and logarithmic sweep functions; 0.1s-999.9s sweep time; forward, backward and roundtrip sweep directions.

Specification:

- General technical parameters Model:JDS6600-30MHz

- Sine frequency range:0~30MHz

- Square frequency range:0~15MHz

- Triangle frequency range:0~15MHz

- Pulse frequency range:0~6MHz

- TTL digital frequency range:0~6MHz

- Arbitrary wave frequency range:0~6MHz

- Square rise time:≤20nS

- Pulse width adjustment range:60nS-4000S

- Mininum Frequency resolution: 0.01µHz（0.00000001Hz）

- Frequency accuracy :±20ppm

- Frequency stability :±1ppm/3h

- Waveform Characteristic Waveform:Sine, Square, Triangle, Pulse (duty-cycle correction,Pulse width and cycle time adjustable), Arbitrary wave, CMOS, DC level, Half-wave, Full-Wave, Pos-Ladder, Neg-Ladder, Noise, Exponential Rise, Exponential Fall, Tone, Cink pulse, Lorentz Pulse, and 60 kinds user defined waveform.

- Display type:2.4 inch TFT LCD

- Voltage range:DC5V±0.5V

Package Includes:

1 piece JDS6600-30MHz dual-channel DDS Function Signal Generator

1 piece USB Cable

1 piece CD

1 piece power adapter

2 pieces BNC to alligator test cable

#### Koolertron 60MHz DDS Signal Generator Counter Upgraded, 2.4in Screen Display High Precision Dual-channel Arbitray Waveform Generator Frequency Meter 200MSa/s (60MHz)

Arbitray Waveform Generator adopts large scale FPGA integrated circuit and high speed MCU microprocessor. The internal circuit adopts active crystal oscillator as benchmark. So the signal stability is greatly strengthened..

Using Dual-channel DDS signal and TTL electric level output to generate precise, stable, low distortion output signal. includes Sine wave, Square wave, Triangle wave, Saw tooth wave, Pulse wave, white noise, user-defined waveform etc. each channel can be independently set the parameters..

With linear sweep(Max. up to 999.9s) and logarithmic frequency sweep functions.Has a frequency measurement, period measurement, positive and negative pulse width measurement and counting function..

Storage feature: You can store 99 groups instrument state parameters set by the user, can be called up to Reproduce. Frequency output of Sine wave can be up to 60MHz. 200MSa/s sampling rate. It has 60 positions for saving user-defined waveform. Waveform Length of each one is 2048 and vertical resolution is 14 bits.

This Signal Generator is the ideal instrument for electronic engineering, laboratories, production lines, teaching and scientific research..

Advantage:

High frequency precision up to 10-6 orders of magnitude.

Full range of 0.01μHz frequency resolution.

Minimum amplitude resolution can be up to 1mV.

Duty cycle of each channel can be adjusted separately. Precision can be 0.1%.

Frequency Characteristic:

Sine frequency range: 0~60MHz

Square & Triangle Wave: 0~15MHz

Pulse & Arbitrary & TTL Digital Signal Wave: 0~6MHz

Square rise time: ≤15nS

Pulse width adjustment range: 30nS~4000S

Frequency accuracy: ±20ppm×10-6

Frequency stability: ±1ppm×10-6/3h

Waveform Characteristic:

Waveform: Sine, Square, Triangle, Pulse (duty-cycle correction), Partia Sine, CMOS, DC level, Half-wave, Full-Wave, Pos-Ladder, Neg-Ladder, Noise, Exponential Rise, Exponential Fall, Tone, Sinc Pulse, Lorentz Pulse, and 60 kinds user defined waveform.

Output Characteristics:

Amplitude resolution: 1mV

Amplitude stability: ±0.5%/5 h

Amplitude flatness: ±5%(10MHz)

Frequency Meter Function:

Frequency measurement range: 1Hz~100MHz

Measurement accuracy: Gate time continuously adjusted between 0.01S~10s

Counter Function:

Counting region: 0-4294967295

Coupled mode: 2kinds Coupling modes, DC and AC

Counting mode: Manual operation

Package Included:

1 x DDS Signal Generator

1 x USB Cable

1 x Power Adapter

2 x Q9 Cable

1 x CD

#### JDS6600 Series 30MHZ Digital Contro Dual Channel DDS Signal Generator/ Pulse Signal Source Frequency Meter

Adopt 320*240 2.4" TFT color LCD display, simultaneously display parameters of CH1 and CH2，Ability to reproduce patterns of custom waveform..

60 groups of arbitrary wave storage function, memory depth of each group is 2048*14bits..

Duty-cycle of each channel can be adjusted independently0.1%-99.9%, the adjusting resolution is 0.1%..

Scanning features: linear sweep and logarithmic sweep functions; 0.1s-999.9s sweep time; sorward, backward and roundtrip sweep directions..

With output short-circuit protection: all signal output can work 60s under load short-circuit condition.

JDS6600 Series 30MHZ Digital Contro Dual Channel DDS Signal Generator/ Pulse Signal Source Frequency Meter

Features:

1.Use soft silicone keys instead of traditional patch keys, high sensitivity and convenient for operation.

2.Using DDS direct digital synthesis technology to generate precise, stable and low distortion output signal.

3.Max. output frequency(sine wave) up to 30MHz, 266MSa/s sampling rate and 14bits vertical resolution.

4.Completely independent dual channels output, synchronizing operation and adjustable phase difference.

5.Great improvement for square wave performance, square frequency range up to 15MHzwith no distortion.

6.High frequency accuracy and resolution: frequency accuracy up to 10^(-6) magnitude, total range frequency resolution is 0.01uHz.

7.High amplitude resolution: the minimum amplitude resolution can reach 1mV, amplitude range is 0Vpp~20Vpp.

8.Pulse pulse width and pulse cycle time is precisely adjustable, adjust the range of 30nS-4000S

9.With pulse trains burst output function, allow the unit output 1~1048575 arbitrary pulse trains.

10.Measurement functions: can measure signal frequency, cycle, positive/negative pulse width and duty cycle; 1Hz~100MHz frequency range.

11.-9.99V~9.99V DC offset range, offset resolution: 0.01V.

12.With digital signal output function, CMOS output range 0~10V.

13.Storage feature: can store 100 groups of parameters set by the user and extract at any time.

14.Counter function: with DC and AC coupling modes.

15.Arbitrary wave editing function: user can edit arbitrary wave on PC and then download to output from the instrument.

16.Communication function: the PC can be used to control the instrument.

Package:

1 x JDS6600 Series 30MHZ Digital Contro Dual Channel DDS Signal Generator/ Pulse Signal Source Frequency Meter

#### JDS6600 Series Digital Control Dual-channel DDS Function Signal Generator Frequency Meter Arbitrary Sine Waveform(40MHz JDS6600-40M)

Using DDS direct digital synthesis technology to generate precise, stable and low distortion output signal.

60 groups of arbitrary wave storage function, memory depth of each group is 2048*14bits.

High frequency accuracy and resolution: frequency accuracy up to 10(-6) magnitude, total range frequency resolution is 0.01uHz.

Use soft silicone keys instead of traditional patch keys, high sensitivity and convenient for operation.

Max. output frequency(sine wave) up to 15MHz/30MHz/40MHz/50MHz/60MHz(optional), 266MSa/s sampling rate and 14bits vertical resolution.

Description:

Adopt 320*240 2.4" TFT color LCD display, simultaneously display parameters of CH1 and CH2.

Silicone button and jog dial,Completely change the previous patch button jerky feeling, in improving the service life and bring you a comfortable experience.

A small signal capable of outputting 2mVpp,The range resolution is 1mV,And you can do 1mVpp,10mVpp, 100MVpp and so on are differentStep value, amplitude range 2mVpp-20Vpp, stepless regulation.

Sinusoidal frequency range is0.01uHz~60MHz,The frequency range of square wave and triangle wave is0.01uHz~15MHz,The frequency range of other waveforms is0.01uHz~6MHzThe frequency accuracy can reach the magnitude of 10-6, and the full range frequency resolution0.01uHzThe accuracy of the output frequency is fully guaranteed;14 bit waveform vertical resolution, 266MSa/s waveform sampling rate, waveform more delicate.

Measurement functions: can measure signal frequency, cycle, positive/negative pulse width and duty cycle; 1Hz~100MHz frequency range.

Using DDS direct digital synthesis technology to generate precise, stable and low distortion output signal.

#### JDS6600 Series Digital Control Dual-channel DDS Function Signal Generator Frequency Meter Arbitrary Sine Waveform(30MHz JDS6600-30M0

Using DDS direct digital synthesis technology to generate precise, stable and low distortion output signal.

60 groups of arbitrary wave storage function, memory depth of each group is 2048*14bits.

High frequency accuracy and resolution: frequency accuracy up to 10(-6) magnitude, total range frequency resolution is 0.01uHz.

Use soft silicone keys instead of traditional patch keys, high sensitivity and convenient for operation.

Max. output frequency(sine wave) up to 15MHz/30MHz/40MHz/50MHz/60MHz(optional), 266MSa/s sampling rate and 14bits vertical resolution.

Description:

Adopt 320*240 2.4" TFT color LCD display, simultaneously display parameters of CH1 and CH2.

Silicone button and jog dial,Completely change the previous patch button jerky feeling, in improving the service life and bring you a comfortable experience.

A small signal capable of outputting 2mVpp,The range resolution is 1mV,And you can do 1mVpp,10mVpp, 100MVpp and so on are differentStep value, amplitude range 2mVpp-20Vpp, stepless regulation.

Sinusoidal frequency range is0.01uHz~60MHz,The frequency range of square wave and triangle wave is0.01uHz~15MHz,The frequency range of other waveforms is0.01uHz~6MHzThe frequency accuracy can reach the magnitude of 10-6, and the full range frequency resolution0.01uHzThe accuracy of the output frequency is fully guaranteed;14 bit waveform vertical resolution, 266MSa/s waveform sampling rate, waveform more delicate.

Measurement functions: can measure signal frequency, cycle, positive/negative pulse width and duty cycle; 1Hz~100MHz frequency range.

Using DDS direct digital synthesis technology to generate precise, stable and low distortion output signal.